This test sequence demonstrates SoundCheck’s Triggered Record – Chirp Trigger function for open loop testing of devices without analog inputs such as smart speakers, wearables, smart home devices, tablets and cellphones. A stimulus WAV file is created in SoundCheck and transferred to the device under test, where it is played back and the response recorded in SoundCheck as if the stimulus were played directly from SoundCheck. The Acquisition step is triggered by the chirp in the stimulus file. Chirp triggers are more robust than level and frequency triggers which are susceptible to false triggering due to background noise.
This sequence demonstrates the two most common microphone measurements, frequency response and sensitivity, on a microphone embedded in a recording device. Typically when measuring a microphone the response of the device can be captured simultaneously with the stimulus. However, with devices such as voice recorders and wireless telephones forming a closed loop can be cumbersome or impossible. This sequence demonstrates how to measure such a device by recording the signal on the device under test, transferring that recording to the computer running SoundCheck and then using a Recall step to import the recorded waveform and analyze it.
Note that this specific sequence, v3, is an improvement on the prior versions. The v1 release required that the audio file containing the recorded response waveform be manually windowed outside of SoundCheck before being analyzed. The v2 release utilized a new feature in SoundCheck 14, using values from the memory list to semi-automatically trim the waveform before analysis. This v3 release completely automates waveform editing.