This sequence is used to measure the NRR, or Noise Reduction Rating, of a hearing protection device to the ANSI S3.19-1974 standard. NRR is a numerical representation of the sound attenuation of a device. The sequence first measures the response spectrum of the unoccluded hearing protector test fixture, then makes a second measurement with the hearing protection DUT affixed. A signal generator virtual instrument generates the pink noise stimulus while an RTA virtual instrument simultaneously records the A and C weighted noise spectrums. The unoccluded and occluded measurements are analyzed with a series of post-processing steps according to the ANSI S3.19-1974 standard. The final display shows the NRR numerical value, RTA spectra of the left and right side of the unoccluded and occluded hearing test fixture, average attenuation level of the DUT, and the standard deviation of the DUT on the test fixture.
This sequence measures the directional response of a microphone and graphs the result as a polar plot. A log sweep stimulus is played from 100 Hz to 10 kHz at each angular increment, and the acquired waveform is analyzed using the Time Selective Response algorithm. This method allows the test to be performed in a non-anechoic environment by placing a window around the direct signal, eliminating the influence of reflections. Commands are sent automatically to the Outline ET250-3D turntable via an ethernet connection, instructing it to move in 10 degree increments after each measurement. The sequence measures the response every 10 degrees from 0 to 180 and mirrors the polar image, which simulates a full 360 degree polar and saves test time. The response at each angular increment is compared against the on-axis response to create a normalized curve. This removes the influence of the device’s frequency response and sensitivity, such that the polar plot only shows the directional response. The final display also contains a graph of the directivity index in decibels versus frequency.