This sequence follows the ANSI S3.22-1996 standard method for testing the frequency response of a hearing aid. An equalized stepped sine sweep from 8 kHz – 200 Hz is played at a level of 60 dBSPL through the anechoic box speaker, and the output of the hearing aid is analyzed with the Heterodyne algorithm to produce a frequency response. Next, the HFA (High Frequency Average) is calculated by averaging the response values at three frequencies (1000, 1600, 2500 Hz). The HFA is then subtracted by 20 dB. Two post processing steps are used to find the upper and lower frequency points at which the response curve intersects this calculated value (HFA – 20 dB). These are the high and low frequency cutoff points.
https://www.listeninc.com/wp/media/logo.png 0 0 Devin Vaillancourt https://www.listeninc.com/wp/media/logo.png Devin Vaillancourt2022-09-12 18:52:192022-09-12 18:52:19Hearing Aid Frequency Response Test Sequence