Filter By Category
IEC-60268-7: Sound System Equipment – Part 7: Headphones and Earphones is an international standard intended to characterize the performance of headphones and earphones. The standard itself is a lengthy document, 9 Sections and 3 Annexes covering 46 printed pages. These SoundCheck sequences focus on the electro-acoustic tests which are detailed in Section 8 “Characteristics to be specified and their method of measurement”.
Five separate sequences are provided, each designed to measure specific characteristics. This approach provides the user with the flexibility to measure all or some of the characteristics of their headphone.
This sequence allows you to test devices without an analog input such as tablets, cellphones and MP3 players. A stimulus WAV file is created in SoundCheck, and copied to the device under test, where it is played and the response recorded in SoundCheck as if the stimulus were played directly from SoundCheck. The stimulus WAV file to be used on the device under test (DUT) may be customized in the stimulus step.
Note that this sequence uses the level-based trigger available in SoundCheck 16.0 and earlier. If you are using version 16.1 or later, please see the frequency-trigger based sequence which takes advantage of new functionality to offer more robust triggering.
This sequence follows the test standard detailed in EN50332-1 (2013) for measuring the maximum sound pressure for portable music players and the earphones/headphones they are bundled with. The test involves loading a weighted pink noise stimulus file (as specified by IEC 268) onto the portable device and playing it through the earphones at the player’s maximum volume.
This sequence performs two of the measurements from the ANSI hearing aid test standard S3.22-1996. The first part, in accordance with section 6.7 of the standard, helps the user set the reference test gain for the hearing aid, which is used for multiple measurements in the standard. The second part, from section 6.12, tests the equivalent input noise (EIN) of the device.
The Hearing Aid Standard suite of sequences contains all the major tests from ANSI S3.22 and IEC 60318-7, as well as some of the additional ‘Annex C’ sequences from the 2003 ANSI standard. Each test, as outlined in the standards, is contained in a single test sequence. These sequences can each be run independently. There are also several full test sequences, which call up the required individual tests to run a complete standard test suite.
These two test sequences are designed to test the hearing aid compatibility of a telephone according to the TIA 1083 and FCC Part 68.316 standards. These sequences, which rely on the user having radial and axial HAC probes, play the appropriate test signals, measure the response and compare them to the limits in the standard. These pre-written sequences enable the user to immediately test to the relevant standards, and can be fully modified to meet any custom testing needs if desired.