This sequence demonstrates an alternative to the traditional SoundCheck single channel impedance measurement method. A stepped sine sweep from 20 Hz to 20 kHz is played through the speaker while the signal across the loudspeaker terminals is recorded by Direct In 1 and the signal across the sense resistor (impedance box) is recorded by Direct In 2. A heterodyne analysis step is then applied to calculate the fundamental response from both inputs and a math post-processing step divides Fundamental A (speaker terminal voltage) by Fundamental B (voltage across sense resistor). A post-processing step corrects for the value of the reference resistor before displaying the final impedance curve. The curve is then post-processed to calculate resonance frequency, maximum impedance and Q of the resonance peak. A set of arbitrary limits steps are also provided to generate pass/fail results.
https://www.listeninc.com/wp/media/logo.png 0 0 Zarina https://www.listeninc.com/wp/media/logo.png Zarina2015-03-11 17:57:582022-01-18 18:47:30Impedance Measurement – Dual Channel Method Using Math Post-Processing